The RC-300 is a hand-held tester that is the answer to portability in a professionally accurate reserve capacity battery load tester!
This premium SLA battery load tester is perfect for alarm service technicians, IT workers, and all wheelchair and scooter technicians. Immediately determines Amp Hours "remaining" in battery
The RC-300 is a portable full-featured menu-driven battery tester that provides quick, professional load results using Auto Meter's Proven Load. Quick and accurate battery load test will increase charging productivity and avoid shut down.
The RC-300 is user friendly. It tells you what to do and provides quick accurate results.
Detailed test results are LCD displayed after each test and can be reviewed and/or printed from memory (printer is optional).
The RC-300 performs a preliminary check on 6, 12, 24 Volt Batteries, load tests 6 Volt and 12 Volt AGM and Gel type batteries. The RC-300 automatically identifies the appropriate voltage and displays the menu selection and instructions needed for the battery being checked or tested.
Applications: 6 & 12 Volt Alarm, Scooter and Wheelchair type Batteries.
Battery Sizes: Load tests batteries rated from 4 to 50 Amp Hours
Load: Automated 40 Amp resistive load
Volts: Measures 0 to 30 Volts with .01 Volt resolution
Memory: Holds Last 50 Tests
Display: Digital 1" x 2.5" - 4 line x 16 character LCD display. Displays menus and test results.
Temperature: 0° to 120° F.
Humidity: 0 to 85% non condensing
Keyboard: Chemical Resistant lexan covered tactile switch
Numerical Entry: 6 Keys: On/Off, +Up, -Down, Y Enter, N Esc, Print
Cooling: Vented Heat Sink
Load Clamps: 2.5 ft. cable 16 Awg. Insulated Volt and Amp conductors
Internal Battery: 9 Volt alkaline
Size: 4.5" x 7.1" x 1.4"
Weight: 1.3 lbs.
Printer: AC-15 infrared printer controlled from up to 40 feet.
Carrying Case: Optional AC-21 case with cigarette lighter adapter and 1 roll of paper.
Adapter Cable: Optional AC-12 serial cable for computer download.
Printer and Case: Optional AC-15 infrared printer and AC-21 Case